 HOME>
 Products>
 Electrodynamic Vibration Test Systems>
 iseries (Standard Range)
Standard Range
iseries
 Enhanced system performance extends test capabilities
 Vibration tests have become diversified and specifications have become increasingly strict.
iseries offer a userfriendly lineup with enhanced performance and durability.
Features
 ■Expanded test range: maximum values that the iseries can offer
 ・Maximum acceleration: 1250m/s^{2} ・Maximum velocity: 3.5m/s ・Maximum displacement: 51mmpp
・Maximum loading mass: 1,000kg  ■Patented upper (armature) support system PS Guide
 Parallel Slope Guide is standard
 ■Low noise
 Optimised design of the air intake based on fluid dynamics has reduced the airintake noise.
 ■All models can be directly coupled to a climatic chamber.
Table Insert Pattern (Unit:mm)
Specifications
System Model  i210/ SA1M 
i220/ SA1M 
i230/ SA2M 
i240/ SA3M 
i250/ SA4M 
i250/ SA5M 
i260/ SA7M ^{*6} 


System Specifications 
Frequency Range(Hz)  04000  03300  03000  02600  02500  02500  02600^{*4}  
Rated Force 
Sine(kN)  3  8  16  24  32  40  54  
Random (kN rms)^{*1} 
3  8  16  24  32  40  54  
Shock(kN)  9  16  32  48  64  80  108  
Maximum Acc. 
Sine(m/s^{2})  1000  1250  1250  1200  914  1142  1000  
Random (m/s ^{2} rms) 
700  875  875  840  640  800  700  
Shock (m/s ^{2} peak) 
2000  2500  2500  2400  1828  2284  2000  
Maximum Vel. 
Sine(m/s)  2.2  2.2  2.2  2.2  2.2  2.2  2.2  
Shock (m/s peak) 
2.2  2.2  2.2  2.2  2.2  2.2  2.2  
Maximum Disp. 
Sine (mmpp) 
30  51  51  51  51  51  51  
Maximum Travel(mmpp)  40  60  64  68  68  68  64  
Maximum Load(kg)  120  200  300  400  600  600  1000  
Power Requirements (kVA) ^{*2} 
6.8  16.4  26  36  51  57  83  
Breaker Capacity (A) ^{*3} 
30  60  100  125  200  200  300  
Vibration Generator 
Model  i210  i220  i230  i240  i250  i250  i260  
Armature Mass(Kg)  3  6.4  12.8  20  35  35  54  
Armature Diameter (φmm) 
128  190  200  290  440  440  446  
Allowable Eccentric Moment(N･m) 
160  294  700  850  1550  1550  1550  
Dimensions(mm) W×H×D 
868 x 700 x 458 
1020 x 903 x 550 
1124 x 957 x 860 
1234 x 997 x 890 
1463 x 1187 x 1100 
1463 x 1187 x 1100 
1527 x 1198 x 1100 

Shaker Body Diameter(mm) 
458  550  630  720  860  860  920  
Mass(kg)  350  900  1500  2000  3000  3000  3500  
Power Amplifier 
Model  SA1M i 10 
SA1M i 20 
SA2M i 30 
SA3M i 40 
SA4M i 50 
SA5M i 50 
SA7M i 60 

Maximum Output (kVA) 
5  10  20  30  40  50  64  
Dimensions(mm) W×H×D 
580 x 1750 x 850 
580 x 1750 x 850 
580 x 1750 x 850 
580 x 1750 x 850 
580 x 1950 x 850 
580 x 1950 x 850 
580 x 1950 x 850 

Mass(kg)  240  280  300  410  850  880  1000  
Controller  Vibration Controller  See Vibration Controller K2  
Cooling  Cooling Method  Air cooling  
Blower  Dimensions (mm) W×H×D^{*5} 
386 x 882 x 369 
492 x 1128 x 625 
606 x 1315 x 891 
707 x 1531 x 917 
1057 x 1841 x 1125 
1057 x 1841 x 1125 
1329 x 2141 x 1080 

Mass(kg)  22  70  168  215  375  375  407  
Wattage(kw)  0.4  1.5  3.7  5.5  11  11  15  
Duct Hose Diameter(φ) 
125  125  200  200  250  250  250 
System Model  i220/EM1M  i230/EM2M  i240/EM3M  i250/EM4M  i250/EM5M  i260/EM7M^{*6}  

System Specifications 
Frequency Range(Hz)  03300  03000  02600  02500  02500  02600^{*4}  
Rated Force 
Sine(kN)  8  16  24  32  40  54  
Random (kN rms)^{*1} 
8  16  24  32  40  54  
Shock(kN)  16  32  48  64  80  108  
High Velocity Shock (kN)^{*7} 
10  23  36  49  63  90  
Maximum Acc. 
Sine(m/s^{2})  1250  1250  1200  914  1142  1000  
Random (m/s ^{2} rms) 
875  875  840  640  800  700  
Shock (m/s ^{2}peak) 
2500  2500  2400  1828  2284  2000  
High Velocity Shock (m/s ^{2}peak)^{*7} 
1562  1796  1800  1400  1800  1666  
Maximum Vel. 
Sine(m/s)  2.2  2.2  2.2  2.2  2.2  2.2  
Shock (m/s peak) 
2.2  2.2  2.2  2.2  2.2  2.2  
High Velocity Shock (m/s peak)^{*7} 
3.5  3.5  3.5  3.5  3.5  3.5  
Maximum Dsp. 
Sine(mmpp)  51  51  51  51  51  51  
High Velocity Shock (mmpp)^{*7} 
51  51  51  51  51  51  
Maximum Travel(mmpp)  60  64  68  68  68  64  
Maximum Load(kg)  200  300  400  600  600  1000  
Power Requirements (kVA) ^{*2} 
16.4  26  36  51  57  83  
Breaker Capacity (A) ^{*3} 
60  100  125  200  200  300  
Vibration Generator 
Model  i220  i230  i240  i250  i250  i260  
Armature Mass(kg)  6.4  12.8  20  35  35  54  
Armature Diameter (φmm) 
190  200  290  440  440  446  
Allowable Eccentric Moment(N･m) 
294  700  850  1550  1550  1550  
Dimensions(mm) WxHxD 
1020 x 903 x 550 
1124 x 957 x 860 
1234 x 997 x 890 
1463 x 1187 x 1100 
1463 x 1187 x 1100 
1527 x 1198 x 1100 

Shaker Body Diameter(mm) 
550  630  720  860  860  920  
Mass(kg)  900  1500  2000  3000  3000  3500  
Power Amplifier 
Model  EM1Mi20  EM2Mi30  EM3Mi40  EM4Mi50  EM5Mi50  EM7Mi60  
Maximum Output(kVA)  10  20  29  40  50  64  
Dimensions(mm) W×H×D 
580 x 1750 x 850 
580 x 1750 x 850 
580 x 1750 x 850 
580 x 1950 x 850 
580 x 2100 x 850 
1160 x 1950 x 850 

Mass(kg)  330  350  460  900  930  1400  
Controller  Vibration Controller  See Vibration Controller K2  
Cooling  Cooling Method  Air cooling  
Blower  Dimensions (mm)WxHxD^{*5} 
492 x 1128 x 625 
708 x 1421 x 782 
707 x 1531 x 917 
1294 x 2123 x 838 
1294 x 2123 x 838 
1298 x 2244 x 798 

Mass(kg)  70  140  215  292  292  305  
Wattage(kw)  1.5  3.7  5.5  11  11  15  
Duct Hose Diameter(φ) 
125  200  200  250  250  250 
 *1 Random force ratings are specified in accordance with ISO5344 conditions. Please contact IMV or your local distributor with specific test requirements.
 *2 Power supply: 3phase 200/220/380/400/415V, 50/60 Hz. A transformer is required for other supply voltages.
 *3 In case of 200 V
 *4 Above 2000 Hz, the force rollsof f at a rate of 12 dB/oct.
 *5 Spec described above is in case of 60Hz. Dimension is changed in case of 50Hz.
 *6 E/L is required for exporting the shaker system of over 50 kN Sine force.
 *7 In case of high velocity option
 * The specification shows the maximum system performance. For longduration tests, derating by up to 70% must be applied. Continuous use at maximum levels may cause failure. Please contact IMV if you use more than 70 %.
 * In the case of Random vibration test, please set the test definition of the peak value of acceleration waveform to be operated less than the maximum acceleration of Shock.
 * Frequency range values vary according to sensor and vibration controller.
 * Armature mass and acceleration may change when chamber is comibined.
 * Mass and dimensions may change for CEmarked systems.