 HOME>
 Products>
 Electrodynamic Vibration Test Systems>
 Kseries (High Excitation Force Water Cooled Range)
High Excitation Force Water Cooled Range
Kseries
 High excitation force and silent water cooled system for improving test environment
 Kseries, high excitation force water cooled vibration simulating test systems fully developed by IMV.
Advanced performance from the Kseries will significantly improve the test environment.
Features
 ■Silent system design
 The water cooling system produces neither the intake nor exhaust sounds that an air cooling system emits.
 ■Record of significant accomplishments
 IMV has been developing the most advanced water cooled system.
Specifications
System Model  K030/ SA4MM 
K062/ SA8MM ** 
K080/ SA14HM ** 
K100/ SA18HM ** 
K125/ SA22HM ** 


Frequency Range(Hz)  03000  02500  02500  02500  02500  
System Specifications 
Rated Force 
Sine(kN)  30.8  61.7  80  100  125 
Random (kN rms)^{*1} 
21.5  61.7  80  100  125  
Shock(kN)  61.6  123.4  160  200  250  
Maximum Acc. 
Sine(m/s^{2})  1000  1000  1000  1000  1000  
Random (m/s^{2} rms) 
557  700  700  700  700  
Shock (m/s^{2} peak) 
2000  2000  2000  2000  2000  
Maximum Vel. 
Sine(m/s)^{*3}  1.8  2.0  2.0  2.0  2.0  
Shock (m/s peak) 
1.8  2.0  2.0  2.0  2.0  
Maximum Dsp. 
Sine(mmpp)  51  51  51  51  51  
Maximum Travel (mmpp) 
58  60  59  62  62  
Maximum Load(kg)  500  1000  1000  2000  2000  
Power Requirements (kVA) ^{*2} 
49  87  100  150  170  
Breaker Capacity (A) ^{*4} 
175  350  350  600  600  
Vibration Generatoer 
Model  K030  K060  K080  K125  K125  
Armature Mass(Kg)  27  40  60  80  80  
Armature Diameter (φmm) 
320  400  446  560  560  
Allowable Eccentric Moment(N･m) 
980  980  1550  2450  2450  
Dimensions(mm) W×H×D 
1100x 1090x 824 
1380x 1085x 1000 
1595x 1050x 1200 
1776x 1373x 1300 
1776x 1373x 1300 

Shaker Body Diameter(mm) 
760  900  1000  1100  1100  
Mass(kg)  3000  3700  5000  7000  7000  
Power Amplifier 
Model  SA4MM K30 
SA8MM K60 
SA14HM K80 
SA18HM K125 
SA22HM K125 

Maximum Output(kVA)  33  60  100  98  124  
Dimensions(mm) W×H×D 
580x 1950x 850 
1160x 1950x 850 
1740x 1950x 850 
2320x 1950x 850 
2320x 1950x 850 

Mass(kg)  950  1350  1500  2500  2600  
Controller  Vibration Controller  See Vibration Controller K2  
Cooling  Cooling Method  Shaker:Water cooling/Power Amplifier:Air Cooling  
Primary Cooling Water Supply （ℓ/min) 
195  260  390  390  390  
Heat Exchanger 
Dimensions (mm) W×H×D 
580x 1700x 850 
580x 1700x 850 
580x 1700x 850 
580x 1700x 850 
580x 1700x 850 

Mass(kg)  400  400  400  400  400 
System Model  K100LS/ SA22HM ** 
K125LS/ SA27HM ** 
K160/ SA26HM ** 
K200/ SA32HM ** 
K350/ SA50HM ** 


Frequency Range(Hz)  02000  02000  02000  02000  02000  
System Specifications 
Rated Force 
Sine(kN)  100  125  160  200  350 
Random (kN rms)^{*1} 
100  125  160  200  315  
Shock(kN)  200  250  320  400  700  
Maximum Acc. 
Sine(m/s^{2})  1000  1000  800  1000  1000  
Random (m/s^{2} rms) 
700  700  560  700  700  
Shock (m/s^{2} peak) 
2000  2000  1600  2000  2000  
Maximum Vel. 
Sine(m/s)^{*3}  2.0  2.0  2.0  2.0  2.0  
Shock (m/s peak) 
2.0  2.0  2.4  2.4  3.5  
Maximum Dsp. 
Sine(mmpp)  100  100  76.2  76.2  76.2  
Maximum Travel (mmpp) 
132  132  86  86  94  
Maximum Load(kg)  2000  2000  2000  2000  3000  
Power Requirements (kVA) ^{*2} 
170  190  270  300  420  
Breaker Capacity (A) ^{*4} 
600  700        
Vibration Generatoer 
Model  K125LS  K125LS  K200  K200  K350  
Armature Mass(Kg)  100  100  200  200  350  
Armature Diameter (φmm) 
560  560  650  650  760  
Allowable Eccentric Moment(N･m) 
2450  2450  4900  4900  4900  
Dimensions(mm) W×H×D 
1990x 1546x 1370 
1990x 1546x 1370 
2465x 1908x 1740 
2465x 1908x 1740 
3020x 2306x 2080 

Shaker Body Diameter(mm) 
1100  1100  1260  1260  1630  
Mass(kg)  8000  8000  16000  16000  37000  
Power Amplifier 
Model  SA22HM K125LS 
SA27HM K125LS 
SA26HM K200 
SA32HM K200 
SA50HM K350 

Maximum Output(kVA)  124  155  256  320  400  
Dimensions(mm) W×H×D 
2320x 1950x 850 
2900x 1950x 850 
3480x 1950x 850 
3480x 1950x 850 
4640x 1950x 850 

Mass(kg)  2600  3300  4850  5000  6200  
Controller  Vibration Controller  See Vibration Controller K2  
Cooling  Cooling Method  Shaker:Water cooling/Power Amplifier:Air Cooling  
Primary Cooling Water Supply （ℓ/min) 
390  390  650^{*5}  650^{*5}  450  
Heat Exchanger 
Dimensions (mm) W×H×D 
580ｘ 1700ｘ 850 
580ｘ 1700ｘ 850 
1050x 1900x 800 
1050x 1900x 800 
1150ｘ 1600ｘ 800 

Mass(kg)  400  400  600  600  600 
System Model  EMK0301  EMK0622 **  EMK0801 **  EMK1251 **  

System Specifications 
Frequency Range(Hz)  03000  02500  02500  02500  
Rated Force 
Sine(kN)  30.8  61.7  80  100  
Random (kN rms)^{*1} 
21.5  61.7  80  100  
Shock(kN)  61.6  123.4  160  200  
Maximum Acc. 
Sine(m/s^{2})  1000  1000  1000  1000  
Random (m/s^{2} rms) 
557  700  700  700  
Shock (m/s^{2} peak) 
2000  2000  2000  2000  
Maximum Vel. 
Sine(m/s)^{*3}  1.8  2.0  2.0  2.0  
Shock (m/s peak) 
1.8  2.0  2.0  2.0  
Maximum Dsp. 
Sine (mmpp) 
51  51  51  51  
Maximum Travel (mmpp) 
58  60  59  62  
Maximum Load(kg)  500  1000  1000  2000  
Power Requirements (kVA) ^{*2} 
49  87  100  150  
Breaker Capacity (A) ^{*4} 
175  350  350  600  
Vibration Generatoer 
Model  K030  K060  K080  K125  
Armature Mass(kg)  27  40  60  80  
Armature Diameter (φmm) 
320  400  446  560  
Allowable Eccentric Moment(N･m) 
980  980  1550  2450  
Dimensions(mm) W×H×D 
1100x 1090x 824 
1380x 1085x 1000 
1595x 1050x 1200 
1776x 1373x 1300 

Shaker Body Diameter(mm) 
760  900  1000  1100  
Mass(kg)  3000  3700  5000  7000  
Power Amplifier 
Model  EM4MM K30 
EM8MM K60 
EM14HM K80 
EM18HM K125 

Maximum Output (kVA) 
33  60  100  98  
Dimensions(mm) W×H×D 
580x 1950x 850 
1160x 1950x 850 
1740x 1950x 850 
1740x 1950x 850 

Mass(kg)  950  1350  1500  2500  
Controller  Vibration Controller  See Vibration Controller K2  
Cooling  Cooling Method  Shaker:Water Cooling/Amp:Air Cooling  
Primary Cooling Water Supply (ℓ/min) 
195  260  390  390  
Heat Exchanger 
Dimensions (mm) W×H×D 
580x 1700x 850 
580x 1700x 850 
580x 1700x 850 
580x 1700x 850 

Mass(kg)  400  400  400  400 
System Model  EMK1252 **  EMK1255 **  EMK1256 **  EMK2001 **  EMK2002 **  

System Specifications 
Frequency Range(Hz)  02500  02000  02000  02000  02000  
Rated Force 
Sine(kN)  125  100  125  160  200  
Random (kN rms)^{*1} 
125  100  125  160  200  
Shock(kN)  250  200  250  320  400  
Maximum Acc. 
Sine(m/s^{2})  1000  1000  1000  800  1000  
Random (m/s^{2} rms) 
700  700  700  560  700  
Shock (m/s^{2} peak) 
2000  2000  2000  1600  2000  
Maximum Vel. 
Sine(m/s)^{*3}  2.0  2.0  2.0  2.0  2.0  
Shock (m/s peak) 
2.0  2.0  2.0  2.4  2.4  
Maximum Dsp. 
Sine (mmpp) 
51  100  100  76.2  76.2  
Maximum Travel (mmpp) 
62  132  132  86  86  
Maximum Load(kg)  2000  2000  2000  2000  2000  
Power Requirements (kVA) ^{*2} 
170  170  190  270  300  
Breaker Capacity (kVA) ^{*4} 
600  600  700      
Vibration Generatoer 
Model  K125  K125LS  K125LS  K200  K200  
Armature Mass(kg)  80  100  100  200  200  
Armature Diameter (φmm) 
560  560  560  650  650  
Allowable Eccentric Moment(N･m) 
2450  2450  2450  4900  4900  
Dimensions(mm) W×H×D 
1776x 1373x 1300 
1990x 1546x 1370 
1990x 1546x 1370 
2465x 1908x 1740 
2465x 1908x 1740 

Shaker Body Diameter(mm) 
1100  1100  1100  1260  1260  
Mass(kg)  7000  8000  8000  16000  16000  
Power Amplifier 
Model  EM22HM K125 
EM22HM K125LS 
EM27HM K125LS 
EM26HM K200 
EM32HM K200 

Maximum Output (kVA) 
124  124  155  256  320  
Dimensions(mm) W×H×D 
2320x 1950x 850 
2320x 1950x 850 
2900x 1950x 850 
3480x 1950x 850 
3480x 1950x 850 

Mass(kg)  2600  2650  3350  4850  5000  
Controller  Vibration Controller  See Vibration Controller K2  
Cooling  Cooling Method  Shaker:Water Cooling/Amp:Air Cooling  
Primary Cooling Water Supply (ℓ/min) 
390  390  390  650^{*5}  650^{*5}  
Heat Exchanger 
Dimensions (mm) W×H×D 
580x 1700x 850 
580ｘ 1700ｘ 850 
580ｘ 1700ｘ 850 
1050x 1900x 800 
1050x 1900x 800 

Mass(kg)  400  400  400  600  600 
 *1 Random force ratings are specified in accordance with ISO5344 conditions. Please contact IMV or your local distributor with specific test requirements.
 *2 Power supply: 3phase 200/220/380/400/415V, 50/60 Hz. A transformer is required for other supply voltages.
 *3 If the tests (Sweep or Spot) include high velocity, the maximum velocity value should be reduced to 1.4 m/s.
 *4 In case of 200 V
 *5 Bypass circuit is needed. Please contact IMV or your local distributor for further information.
 * The specification shows the maximum system performance. For longduration tests, derating by up to 70% must be applied. Continuous use at maximum levels may cause failure.
 * In the case of Random vibration test, please set the test definition of the peak value of acceleration waveform to be operated less than the maximum acceleration of Shock.
 * Frequency range values vary according to sensor and vibration controller.
 * Mass and dimensions may change for CEmarked systems.
 ** E/L is required for exporting the shaker system of over 50 kN Sine force.